The HI7061L is a general-purpose cleaning solution for use with pH and ORP electrode that can be used for a variety of applications.
Use our General Purpose pH Electrode Cleaning Solution to prolong the life of your electrode and ensure accurate readings.
About This Item
Hanna General Purpose pH Electrode Cleaning Solution (500 mL):
Specially formulated for cleaning pH and ORP electrodes
Clearly marked expiration date and lot number for traceability.
Air-tight seal to ensure the quality of the solution
The most common cause for pH measurement inaccuracies is an unclean or improperly cleaned electrode. This is very important to note because, during calibration, the instrument assumes that the electrode is clean and that the standardization curve created during the calibration process will remain a valid reference until the next calibration. pH meters on the market today will allow an offset voltage of approximately ±60 mV. The deviation from 0 mV is not unusual but ideally should be no greater than ±30 mV. The calibration process compensates for the change in offset voltage. If the mV offset continues to deviate and the meter is calibrated with a dirty electrode then inaccurate readings will result.
Cleaning is a fast and effective routine that should be performed on a regular basis as a preventative measure against using a dirty electrode and to ensure that the junction is not clogged. By simply immersing the electrode in the HI7061L cleaning solution for 15 to 20 minutes will dissolve any mineral deposits or other coatings.